The results of the COM 2 loop back test are displayed.
Press Esc or F6 Cancel when the test is complete. If you connect to an inactive network tap, the test result may be a false negative. Select the LAN Test button. Press the Enter key to select the Test network connectivity button.
The Checking connectivity. Please wait. Then, the test results are displayed.
Press Esc or F6 Cancel after the test results are displayed. Connect the MAC device to an active analog phone line. NOTE Ensure that the phone line is active. If you connect to an inactive phone line, the test result may be a false negative. Select the Modem Test button.
Test the internal modem as described in the following steps. With the focus on the Internal Modem Test button, press the Enter key. The Test in Progress. Please wait message is displayed.
Then the results of the test are displayed. If the system has an external modem, go to step 5. If the system has only the internal modem option, press F6 to select Cancel. Then, go to step 6. Test the external modem as described in the following steps. With the focus on the External Modem Test button, press the Enter key. If the External Modem Test failed, replace the external modem. Select the USB Test button. Press F4 to select Yes if the test passed or press F5 to select No if the test failed.
Connect a patient cable with lead wires to the MAC patient cable connector. Connect all leads to a patient simulator or shorting bar. Select the Patient Lead Wire Check button. Press the Enter key to select the Start Test button. For each lead wire, the test results are displayed. Replace every lead wire that failed the test.
Repeat the test. Careful skin preparation is the key to an interference-free ECG.
Rhythm report is not stored and cannot be transmitted. Plastics kit, PN Table It can be used as a reference if system setup needs to be restored manually. For instructions on inserting the battery, refer to Replacing the Battery on page NOTE Do not use the system on battery power until the battery is fully charged as indicated by the on-screen battery gauge and the blank LED next to the display. Reattach the three cables A shown in the following figure. Devices may be connected to other devices or to parts of systems only after making certain that there is no danger to the patient, the operators, or the environment as a result. Not only must ECG systems be portable enough to easily reach your patients anywhere in the world, but ECG data also must be able to travel, too.
Signal quality is indicated using Hookup Advisor. Hookup Advisor can be turned on or off in the ECG menu.
However, if the unit is malfunctioning and any of the following error messages are repeating and unrecoverable, replace the FRUs in the order as listed. Acquisition error codes Where multiple solutions are suggested, perform the suggested actions in their listed order. Once a solution restores functionality, do not attempt the next solution in the list. Table 2. Check cables. Self correcting internal error. Reload Application software Acquisition Error -3 Acquisition buffer overflow 1.
Reload application software 2. Replace if necessary. A: Perform the following steps: 1. Push the SD card into slot to seat it in place. Press F6 again to select More. Press F3 to select Export Setup. Select Setup for Export. Highlight the setup file you want to save to SD card from the list on left side of the window. Press F1 to select Export. The Configuration was successfully exported window opens. Press F6 to select OK. Eject the SD card by pushing it in once. Store it to a secure location. Insert the SD card in the SD card slot in the back panel as described previously.
Press F2 to select Import Setup. Highlight the setup file you want to import from the SD card from the list on right side of the window.
Press F1 to select Import. The Configuration was successfully imported window opens.
How do I obtain this report? Press F3 to select Print Setup Report. Move the focus to the Complete Setup button and press the Enter key to select. Save the printed setup report in a secure location. It can be used as a reference if the system setup needs to be restored manually. Where can I find the option codes?
However, these codes will be listed on the last page of your printed setup report. These codes are also found on a label next to the battery compartment. The printed list of option codes is a more convenient aid to use when these codes need to be re-entered after the new mainboard is installed. A: Clean the exterior surfaces of all the equipment and peripheral devices monthly, or more frequently if needed. Do NOT drip water or any liquid on the writer assembly, and avoid contact with open vents, plugs, and connectors. Battery capacity Q: What is the capacity of the battery?
GE MAC ECG Analysis System - Service Manual - Download as PDF File . pdf), Text File .txt) or read online. Manual de servicio. Medical Equipment GE MAC Operator's Manual. Ecg analysis system ( pages). Medical Equipment GE CareScape V Service Manual. Vital sings.
However, if operating the device off AC power, be aware that a fully-charged battery is capable of taking approximately 50 or more ECGs with one-page printed reports or three hours of continuous operation without printing. How do I obtain the MAC address? A: Follow these steps to obtain the MAC address: 1. Move the focus to the Service Report button and press the Enter key to select. Find the MAC address on the printed service report. A: The Location number can be set in Basic Setup to save you from entering it for each test.
Press F1 to select Basic Setup.